The Accu-Chek FastClix lancing device is the world’s first 1-Click lancing device with a drum. Its unique technology makes lancing quick, easy and is virtually pain-free. The rotatable cap offers 11 depth settings that let you customize the device to your skin type. The lancing device uses a drum of 6 preloaded lancets, so you don’t handle individual lancets.Product Manuals
Accu-Chek Fastclix is available in selected countries in the Middle East and Africa region. Please visit our Contact Us page to get in touch with your local Roche Diabetes Care entity or Roche Distributor.
Control penetration with 11 depth settings
0.7 to 2.2 mm
Prime and release with 1-Click action
119 mm x 19 mm
Yes, AST capability is available upon request
The Accu‑Chek FastClix lancing device is intended for personal use only. It is only allowed to be used to obtain blood from one person. There is a risk of infection being transmitted if the lancing device is used by other people, even by family members, or if healthcare professionals use this lancing device to obtain blood from different people. This lancing device is therefore not suitable for professional use in healthcare facilities and institutions.
Uncritical plastic material
Yes. The number of available lancets is indicated on the window next to the lever.
A red mark is displayed on a used drum.
No, not without misuse.
Always use a new lancet when you obtain blood. In doing this, you reduce the risk of infection, and obtaining blood remains virtually pain free.
0.3 mm (30G)
1-Click action enables you to prime and release in one simple action. This makes it quick and more convenient to obtain a blood sample as there are fewer handling steps. Always wash and dry hands before testing.
Inserting a drum
Setting the penetration depth
Turn the rotatable cap to select from 11 settings. Start with 2, and adjust depending on your skin type.
Obtaining a blood sample
Push the release button to prime and release in 1-Click action
Loading a new lancet
There is a lever next to the window showing the number of available lancets. Slide the lever back and forth to change to a new lancet. The number in the window changes, telling you how many lancets remain in the drum.