The Accu-Chek Multiclix lancing device has been the first to incorporate a unique, 6-lancet drum. With no single lancets to see or touch, lancing is safe and convenient. It also features multiple penetration depths. This allows you to choose the shallowest possible penetration depth for your individual skin type and helps to minimize tissue damage.Product Manuals
Control penetration with 11 depth settings
0.7 to 2.2 mm
Prime button to prime and yellow release button
Yes, the number of available lancets is indicated in the window.
A blue ring and red mark is displayed on a used lancet drum.
No, not without misuse.
Always use a new lancet when you obtain blood. In doing this, you reduce the risk of infection, and obtaining blood remains virtually pain-free.
0.3 mm (30G)
Always wash and dry hands before testing.
Inserting a lancet drum
Setting the penetration depth
Turn the rotatable cap to select from 11 settings. Start with 2, and adjust depending on your skin type.
Obtaining a blood sample
Press the priming button to prime. Press the yellow button to release the lancet.
Loading a new lancet
Twist the priming button to change to the next lancet. The numbers on the window show how many lancets are left.